443010, Самара, ул. Галактионовская, 141.
ИНН 6315522386 / КПП 631501001
Р/сч. 40702810954390101485 в Поволжском банке ОАО «Сбербанк России» г. Самара
К/сч. 30101810200000000607, БИК 043601607
From the date of establishing in 1993, Design Bureau “Fizelektronpribor” has implemented more than 80 process automation projects at various Russian industrial plants. A lot of plants including foreign ones use our instruments.
Among our customers:
Design Bureau “Fizelektronpribor” has an experience of process automation in various industries. We have implemented automation control systems in chemical industry, at petroleum plants, building material factories and other productions.
Some of our projects:
In what way calibration of your moisture sensor installed in the sand and gravel bunker is made? (By method of sample drying or by any other)?
Calibration for the hopper and dispenser is performed using results of samples drying. The customer can independently change the calibration using results of his own measurements. Also there is the second way. It is possible to send you a calibration file prepared on the base of data received from you. This file should be installed on the moisture using supplied software and an ordinary PC.
The moisture meter connects to PC via RS485, and an adapter allowing connection of RS485 line to the USB port can be supplied in the package.
Information about calibration is also listed in the moisture meter operation manual.