443010, Самара, ул. Галактионовская, 141.
ИНН 6315522386 / КПП 631501001
Р/сч. 40702810954390101485 в Поволжском банке ОАО «Сбербанк России» г. Самара
К/сч. 30101810200000000607, БИК 043601607
From the date of establishing in 1993, Design Bureau “Fizelektronpribor” has implemented more than 80 process automation projects at various Russian industrial plants. A lot of plants including foreign ones use our instruments.
Among our customers:
Design Bureau “Fizelektronpribor” has an experience of process automation in various industries. We have implemented automation control systems in chemical industry, at petroleum plants, building material factories and other productions.
Some of our projects:
The moisture meter is made according to VIGT.415210.100TU specifications on the base of a set of design documents VIGT.415210.100.
|Moisture measurement range - mass water content, W, % (see note 1)||0 to 100|
|Acceptable absolute error limits Δ, mass water content, % (Δ is calculated according to the formulae depending on W, where W – is the result of moisture measurement in percent– see note 2)||Δ= 0,035+0,05·W|
|Temperature measurement range, °С||-40… 150|
|Temperature range of moisture meter calibration, °С (see note 3)||+5…40|
|Range of sensor operational temperatures, °С||-20…80|
|Range of electronic unit operational temperatures, °С||-20…80|
|Measurement period, sec||0,2|
RS485 Modbus RTU
|Electronic unit supply voltage, V
|Current consumption, mA, not more than||200|
|Sensor overall dimensions, mm||Ø 108х120|
|Sensor weight, kg||3,5|
|Electronic unit dimensions, mm||255 x 170 x 60|
|Electronic unit weight, kg||2|
|Enclosure protection rating of electronic unit according to IEC 60529||IP54|
|Enclosure protection rating of sensor according to IEC 60529||IP67|
|Length of communications cable between sensor and electronic unit, m (see note 4)||1,5…2|
|Maximum allowable length of cable transmitting RS485 digital signal from the electronic unit to an external control device (controller, computer), not less than, m||1000|
|Maximum allowable length of cable transmitting 4-20mА analog signal from the electronic unit to an external indicating device, not less than, m||100|
|Moisture range W||Acceptable absolute error limits Δ|
|up to 10%